Automated handling of up to 23 samples in quarz tubes 3-6 mm diameter, precise height positioning within resonator for highest reproducibility
Automatic liquid sample transfer for spectrometers of the MiniScope series, autosampler and integrated peristaltic pump, software controlled. Software for kinetic measurements, including automatic data acquisition and data evaluation. Single or multiple chemicals, optional: automated component mixing and temperature control.
Suitable for 14 liquid samples and cleaning liquid
Suitable for 23 liquid samples and cleaning liquid
The X-ray unit is used for the simulation of environmental impacts on samples.
This unit consists of,
Automated loading of X-ray source
fully automated 50 kV X-ray tube at a focal spot size of 1mm
high voltage generator
Ionisation chamber for X-ray dose measurements
fully integrated dose measurement for precise monitoring of X-ray irradiation intensity
For the measurement of samples in the EPR/ESR Spectrometer one has the choice between the following sample vessels,
Fixed Temperature dewar
Capillaries are standardized calibrated sample vessels. They are cheap one way items and allow an easy sample exchange. Usually
capillaries are used together with a matching guidance tube. For aqueous solutions the maximum volume of capillaries is 50 µl. In the case you use organic solvents with a lower dielectric
constant, you can use capillaries with a higher volume or sample tubes.
Sample Tubes are used for liquid and solid samples. We offer sample tubes of different diameters and length. The dielectric constant of the solvent of your liquid sample lays down the diameter of the sam- ple tube. For example radicals in hydrocarbons can be measures using a 4 mm or 5 mm sample tube. Concerning the solid samples the humidity of the sample restricts the diameter of the sample tube. The length 135 mm is used in general. The length 240 mm is adapted to the finger dewars, we offer. In the case a sample is not tuneable you can either choose a sample tube with a smaller diameter or you reduce the amount of sample inserted into the resonator.
Flat Cells are used for liquid samples in solvents with a high dielectric constant. The ideal sample distribution in the resonator leads to a 2 – 2.5 fold higher intensity compared to 50 µl capillaries. The crucial point, when using a flat cell is the orientation of the flat section, the vertical position and the filled volume of the flat cell.
Therefore we recommend using the special holder SH-P as the system to achieve always the same sample positioning.
Tissue Cells can be used to measure skin sections, paste and emulsions. The same requirements have to be fulfilled that are valid for flat cells.
A Special Cuvette we offer for alanine dosimetry of thin film test strips. This cuvette has a flat section too. Furthermore we offer special tubes for alanine tablet dosimeters.
Fixed Temperature Dewars are used to measure samples at liquid nitrogen temperature of 77K. You can either drop the sample into the liquid nitrogen and the frozen drops are collected in the finger for mea- surement. A further possibility is to freeze a cylinder of 4 mm diameter matching to the size of the inner finger of the dewar. In the case you like to measure solid samples you can use sample tubes of 3 or 4 mm outer diameter and 240 mm length.